ULTRASOUND FOR PHYSICAL THERAPY 1-3 MHZ DIGITAL SONOMED V CARCI BRAZIL
Ultrasound for physiotherapy and electrotherapy with 1.0 and 3.3 MHz transducers.
Therapeutic resource used for the treatment of the most diverse musculoskeletal disorders (dislocations, fractures, contractures, muscle spasms), acute and chronic inflammatory conditions and aesthetic treatment.
Emission in continuous mode, and Pulsed in the frequencies: 100 Hz, modulated in 50%, 20% and 10%; 48 Hz, modulated at 20% and 10%; 16 Hz, modulated at 20% and 10%.
Memory of 17 treatment protocols and recording of up to 18 user protocols.
Panel with smooth keypad, and graphic blue liquid crystal display with backlight.
Transducer overheating protection.
Detection of bad contact of the transducer cable.
Anatomical and waterproof transducer.
Power: 100 to 230Vac 50/60Hz (automatic) (+/- 10%)
Maximum consumption : 65 VA Fuses : 2.0 A (250V 20AG) Operating Frequencies : 1.0 & 3.3 MHz ± 5% US Output Operating Mode : Continuous and Pulsed Output control indication : Intensity in W /Cm2 or Watts Maximum effective output power: Continuous mode : 7.2 W ± 20% Pulsed mode : 10.5 W ± 20% Maximum effective intensity : Continuous mode : 2 W/Cm2 Pulsed mode : 3 W/Cm2 Timer : 1 to 20 minutes Protection against risk of electric shock: Class II Degree of protection of the applied part: Type BF Protection against water ingress: IPX0. Compliance with NBR IEC 60601.1, NBR IEC 60601.1.2 and IEC 60601.2.5 standards: Mode of operation: Continuous UMDNSTM Rating: 11-248
01 Applicator Heads (transducers): Type : Collimated Operating Frequency : 1.0 & 3.3 MHz ± 5% Geometric Area : 10 Cm2 Effective Radiation Area (ERA) : 3.5 Cm2 ± 20% Intensity Ratio (BNR) Max . : < 6.0 : 1 Water tightness : IPX7
Weight : 215 grams
Maximum Acoustic Power : 10 Watts
This product must be used by a health professional.
The colors shown on the site are merely illustrative and may or may not exactly match the shade applied.
For more information about handling and care regarding the use of the product, consult the User Manual.
Purely illustrative image and product description is the responsibility of the manufacturer.